- Research article
- Open Access
- Open Peer Review
Microsatellite instability testing in colorectal cancer using the QiaXcel advanced platform
BMC Cancer volume 18, Article number: 484 (2018)
Microsatellite instability (MSI) is a major predictive and diagnostic marker in several cancers including colorectal carcinomas. Diagnostic testing for microsatellites is generally performed using capillary sequencers, which requires expensive high-end equipment including expensive chemistry using fluorescent dyes labelling the PCR products of interest. In this study we have modified such a diagnostic protocol and established the microsatellite testing on the QiaXcel Advanced platform.
MSI testing was based on a previously established protocol describing a multiplex PCR followed by fluorescent detection of PCR products in a capillary sequencing device. Ten microsatellites were included in the new protocol: BAT25, BAT26, BAT40, D2s123, D10s197, D13s153, D17s250, D18s58, D5s346, and MycI. In this protocol the PCR was demultiplexed and established on the QiaXcel Advanced system (Qiagen, Hilden, Germany).
Making use of a series of FFPE control samples with known MSI status including those with and without MSI a protocol for MSI testing was successfully established on the QiaXcel Advanced platform.
MSI testing for human colorectal cancers using the QiaXcel Advanced system could serve as an economic acceptable tool for rapid diagnostics in laboratories that do not have access to a capillary sequencing unit.
Microsatellites are non-coding DNA sequences that occur ubiquitous in all eukaryotic genomes and are a powerful tool for the analysis of populations, genetic diversity, and paternity tests . The analysis of microsatellites therefore is used in many disciplines, including botany, genetics, zoology, medical microbiology, and others [1,2,3,4,5,6,7,8,9].
Also pathologists and oncologists have become aware of microsatellites, as microsatellite instabilities (MSI) frequently occur in several human cancers, mainly in colorectal carcinomas. These instabilities indicate that the mismatch repair system of the host cells is likely damaged and can serve as a predictive and diagnostic marker [10,11,12,13,14].
As early as in 2005 Popat and coworkers have published a systematic review of the instability of microsatellites and their usage as markers in the prognosis of colorectal cancers . The authors have concluded that patients with MSI have a significantly better prognosis than MS-stable tumors and had a better response to chemotherapy (reviewed by ). Meanwhile it has been shown that also other tumor types could be associated to MSI . In particular, 14 of 18 cancers, such as endometrial, gastric, and colon cancer had high percentages of MSI. The study by Hause and colleagues  also revealed that MSI testing can be used to classify tumor types on a molecular level in four different groups named A-D; e.g. colon and rectal cancers clustered in group A, whereas liver hepatocellular carcinomas and kidney renal carcinomas clustered in group D. These two studies are examples that clearly show that MSI testing is a useful tool to determine the molecular tumor type and enables the pathologist to make a better prognosis for the subsequent chemo-therapy.
So far the most requested MSI testing in colorectal cancers appears to be the most relevant testing in Germany and since a few years there is also a round robin trial organized by German pathologists. However, diagnostic testing for microsatellites generally requires high-end laboratory equipment in the form of a capillary sequencing device capable to distinguish between multiple fluorescent dyes, thus the MSI testing is limited to highly specialized laboratories. A more broadly available device are capillary electrophoresis systems that are frequently used in diagnostic laboratories for quality analyses of DNA and RNA isolated from clinical specimen including formalin fixed paraffin embedded (FFPE) tissues. The aim of this study was to establish a protocol for MSI testing on the QiaXcel Advanced system (Qiagen, Hilden, Germany). It was previously shown that this system is feasible for plant genotyping via microsatellite analyses  and can be used for MSI testing in human endometric cancers , but was not yet used for the analysis of colorectal tumors.
DNA used for this study was obtained from clinical samples previously tested positive for MSI by immunohistochemistry for mismatch repair defects. Unfortunately, as these samples were tested externally for MSI by IHC, no more details on the results of MSI-IHC testing were available. A collection of 5 samples with MSI and 5 control samples without MSI could be included in this pilot study. In all ten cases healthy tissues of the respective patients were use as controls. DNA was extracted using Maxwell FFPE DNA extraction kits (Promega, Mannheim, Germany). In addition, we have included 8 samples from a recent German round robin trial for HPNCC MSI (http://www.quip-ringversuche.de/pdf/2017/QuIP-Programm-2017.pdf; last page view 16th April 2018). For these samples we have received the results for MSI testing for markers of the Bethesda panel and the information if the samples had a MSI-high, MSI-low or MS-stable status.
For this study a previously established MSI detection protocol that included the following microsatellites has been de-multiplexed (Table 1). This primer set includes the recommend Bethesda protocol primers  plus 5 additional markers that could be useful for MSI detection [20, 21]; this latter extended panel was shown to be useful to confirm the MSS status of hereditary colorectal carcinoma not caused by common mutations in the mismatch repair genes . According to the Bethesda agreement a sample is considered as MSI-H if two or more markers of the Bethesda panel are mutated. Thereby, the additional markers were recommended by Wolfgang Dietmaier (University Hospital Regensburg, personal communication) who organized the last German round robin trial for testing of MSI in hereditary colorectal cancer. The additional markers serve as back-up if one of the Bethesda panel markers cannot be properly determined due to technical reasons such as PCR inhibition or DNA fragmentation.
For the PCR the HotStar Taq PCR Mastermix Kit (Qiagen, Hilden Germany) was used according to the manufacturer’s recommendation with respectively 1 μl forward and reverse primer (10 pmol/μl each). The final volume of each PCR was 25 μl including 5 μl from the extracted DNA (1 ng/μl). Detection and differentiation of PCR fragments were performed on the QiaXcel Advanced System using a run method based on the instrument setting OM500, AM 15 bp–600 bp, and SM 25-500 bp.
Microsatellite testing was performed in a set of clinical controls and round robin trial specimens that were previously tested positive for MSI by immunohistochemistry (IHC) and confirmed in an external laboratory by fluorescent dye based capillary electrophoresis..Based on these specimen cohort PCRs were performed for the 10 microsatellites BAT25, BAT26, BAT40, D2s123, D10s197, D13s153, D17s250, D18s58, D5s346, and MycI, of which BAT25, BAT26, D5s346, D17s250 and D2s123 originate from the Bethesda panel [10, 12, 16, 22]. In total 20 PCRs were performed per patient, as for every microsatellite tumor DNA was compared to DNA from the healthy control tissue, respectively.
In case of stable microsatellites, the electropherogram shows the same pattern in healthy tissue as well as in the tumor, but may vary in the overall intensity (Fig. 1).
Unstable microsatellites have been identified in all previous MSI positive tested samples with the QiaXcel system in direct comparison between healthy tissue and tumor as exemplarily shown for BAT26, D13s153, D18s58 (Fig. 2). Figure 3 shows the comparison of healthy versus tumor tissue for all 10 microsatellite markers included in our protocol. In most cases the differences are obvious and result in a different peak pattern in case of the instabilities, mostly characterized by additional peaks or a peak-shift to the right side of the diagram (which in turn indicates longer MS-sequences).
The re-testing of round robin trial samples revealed a full match between the round robin results that were generated with a capillary sequencing unit and the protocol established for the present study (Table 2). Thereby, the round robin trial was designed according to the Bethesda panel and it was not discriminated between MSI-low (MSI-L) and MS-stable (MSS). However, all unstable tumors were MSI-high (MSI-H) in the round robin samples.
The QiaXcel system was previously used for microsatellite analyses for plant species and for endometrial cancer [1, 18], but so far has not been used for colorectal cancer typing. In our pretested clinical pilot cohort the QiaXcel advanced system confirmed the previous IHC results of all samples. Because of the small patient cohort we were not able to detect instabilities for BAT25, D10s197, D17s250 and MycI, but were able to detect those instabilities in a further set of samples originating from the last round robin trial we participated (examples shown in Fig. 3). Our results show, that this typing is also possible with the QiaXcel system and it thus may be used in those pathology institute that have not yet established MSI-testing on other platforms or lack a capillary sequencing device.
For the estimation of the assay costs, the reagents used for the proposed method and for usage of a capillary sequencer-based protocol were taken into account. For capillary electrophoresis dye-labeled primers, the capillary usage itself, and the list price for multiplex-suitable PCR chemistry was calculated. Per patient two reactions would be required for the tumor tissue and two for the tumor free tissue, resulting in approx. 17–18 € per patient. The QiaXcel capillary unit costs approx. 750 € plus PCR reagents, and has 104 runs with 12 lanes per run. With the proposed protocol 10 lanes are required for the tumor free tissue, and 10 lanes are used for the tumor tissue, plus an additional lane for the size marker, resulting also in 17–18 € per patient analysis. The “hardware” costs, however, differ significantly. The German list price for the QiaXcel Advanced system is approx. 29.000 €, whereas the list price for a capillary sequencing unit is between 60.000 and 120.000 €.
The only experienced disadvantage during the setup of the MSI testing protocol was that de-multiplexing rapidly leads to large experimental approaches of 20 PCRs per patient being error-prone in routine settings with large numbers of samples. To reduce the amount of single PCRs it would be possible to combine such targets with distinct difference in size. The fact that microsatellites vary among individuals is technically used for phylogenetic analyses such as parental testing [7, 8], but implicates for colorectal cancer testing that for each patient corresponding tumor-free tissue controls are required. And it must be taken into account that variances in the overall intensity of the electropherograms most likely result from DNA quality and DNA preparation from FFPE material.
In summary, we come to the conclusion that the QiaXcel system is appropriate for validation for routine diagnostics of MSI testing in colorectal cancers and routine usage in pathology laboratory without a capillary sequencing device available.
formalin fixed paraffin embedded
Dean DA, Wadl PA, Hadziabdic D, Wang X, Trigiano RN. Analyzing microsatellites using the QIAxcel system. Methods Mol Biol. 2013;1006:223–43.
Manrique P, Hoshi M, Fasabi M, Nolasco O, Yori P, Calderon M, Gilman RH, Kosek MN, Vinetz JM, Gamboa D. Assessment of an automated capillary system for plasmodium vivax microsatellite genotyping. Malar J. 2015;14:326.
Coetzer WG, Downs CT, Perrin MR, Willows-Munro S. Testing of microsatellite multiplexes for individual identification of cape parrots (Poicephalus robustus): paternity testing and monitoring trade. PeerJ. 2017;5:e2900.
Hoekert WE, Neufeglise H, Schouten AD, Menken SB. Multiple paternity and female-biased mutation at a microsatellite locus in the olive ridley sea turtle (Lepidochelys olivacea). Heredity. 2002;89(2):107–13.
Ingvarsson S, Finnsdottir V, Sigurdsson A, Geirsson G. Population studies and validation of paternity determinations by six microsatellite loci. J Forensic Sci. 2000;45(3):692–5.
McVay JD, Rodriguez D, Rainwater TR, Dever JA, Platt SG, McMurry ST, Forstner MR, Densmore LD 3rd. Evidence of multiple paternity in Morelet's crocodile (Crocodylus moreletii) in Belize, CA, inferred from microsatellite markers. J Exp Zool A Ecol Genet Physiol. 2008;309((10):643–8.
Sakaoka K, Suzuki I, Kasugai N, Fukumoto Y. Paternity testing using microsatellite DNA markers in captive Adelie penguins (Pygoscelis adeliae). Zoo biology. 2014;33(5):463–70.
Song SD, Drew RA, Hughes JM. Multiple paternity in a natural population of a wild tobacco fly, Bactrocera cacuminata (Diptera: Tephritidae), assessed by microsatellite DNA markers. Mol Ecol. 2007;16(11):2353–61.
Xue D, Zhang T, Liu JX. Microsatellite evidence for high frequency of multiple paternity in the marine gastropod Rapana venosa. PLoS One. 2014;9(1):e86508.
Bettstetter M, Dechant S, Ruemmele P, Grabowski M, Keller G, Holinski-Feder E, Hartmann A, Hofstaedter F, Dietmaier W. Distinction of hereditary nonpolyposis colorectal cancer and sporadic microsatellite-unstable colorectal cancer through quantification of MLH1 methylation by real-time PCR. Clinical cancer research : an official journal of the American Association for Cancer Research. 2007;13(11):3221–8.
Bettstetter M, Woenckhaus M, Wild PJ, Rummele P, Blaszyk H, Hartmann A, Hofstadter F, Dietmaier W. Elevated nuclear maspin expression is associated with microsatellite instability and high tumour grade in colorectal cancer. J Pathol. 2005;205(5):606–14.
Dietmaier W, Gansbauer S, Beyser K, Renke B, Hartmann A, Rummele P, Jauch KW, Hofstadter F, Ruschoff J. Microsatellite instability in tumor and nonneoplastic colorectal cells from hereditary non-polyposis colorectal cancer and sporadic high microsatellite-instable tumor patients. Pathobiology : journal of immunopathology, molecular and cellular biology. 2000;68(4–5):227–31.
Giuffre G, Muller A, Brodegger T, Bocker-Edmonston T, Gebert J, Kloor M, Dietmaier W, Kullmann F, Buttner R, Tuccari G, et al. Microsatellite analysis of hereditary nonpolyposis colorectal cancer-associated colorectal adenomas by laser-assisted microdissection: correlation with mismatch repair protein expression provides new insights in early steps of tumorigenesis. The Journal of molecular diagnostics : JMD. 2005;7(2):160–70.
Wild PJ, Reichle A, Andreesen R, Rockelein G, Dietmaier W, Ruschoff J, Blaszyk H, Hofstadter F, Hartmann A. Microsatellite instability predicts poor short-term survival in patients with advanced breast cancer after high-dose chemotherapy and autologous stem-cell transplantation. Clinical cancer research : an official journal of the American Association for Cancer Research. 2004;10(2):556–64.
Popat S, Hubner R, Houlston RS. Systematic review of microsatellite instability and colorectal cancer prognosis. Journal of clinical oncology : official journal of the American Society of Clinical Oncology. 2005;23(3):609–18.
Dietmaier W. Microsatellite instability. A new predictive marker (?). Pathologe. 2010;31(Suppl 2):268–73.
Hause RJ, Pritchard CC, Shendure J, Salipante SJ. Classification and characterization of microsatellite instability across 18 cancer types. Nat Med. 2016;22(11):1342–50.
Pecorino B, Rubino C, Guardala VF, Galia A, Scollo P. Genetic screening in young women diagnosed with endometrial cancer. J Gynecol Oncol. 2017;28(1):e4.
Boland CR, Thibodeau SN, Hamilton SR, Sidransky D, Eshleman JR, Burt RW, Meltzer SJ, Rodriguez-Bigas MA, Fodde R, Ranzani GN, et al. A National Cancer Institute workshop on microsatellite instability for cancer detection and familial predisposition: development of international criteria for the determination of microsatellite instability in colorectal cancer. Cancer Res. 1998;58(22):5248–57.
Dietmaier W, Wallinger S, Bocker T, Kullmann F, Fishel R, Ruschoff J. Diagnostic microsatellite instability: definition and correlation with mismatch repair protein expression. Cancer Res. 1997;57(21):4749–56.
Schiemann U, Muller-Koch Y, Gross M, Daum J, Lohse P, Baretton G, Muders M, Mussack T, Kopp R, Holinski-Feder E. Extended microsatellite analysis in microsatellite stable, MSH2 and MLH1 mutation-negative HNPCC patients: genetic reclassification and correlation with clinical features. Digestion. 2004;69(3):166–76.
Dietmaier W, Hofstadter F. Detection of microsatellite instability by real time PCR and hybridization probe melting point analysis. Laboratory investigation; a journal of technical methods and pathology. 2001;81(10):1453–6.
The authors are grateful to Prof. Dr. Wolfgang Dietmaier, Institute for Pathology, University of Regensburg, Germany for providing the multiplex MSI testing protocol and critical comments on the manuscript.
Availability of data and materials
All data of this study are included in the manuscript. Primer sequences are presented in the manuscript; all other materials except clinical specimen are available from the respective vendors.
Ethics approval and consent to participate
This study was performed in agreement with the declaration of Helsinki and in accordance with a vote from the ethical committee of the Private University of Witten/Herdecke (75/2013). Due to this vote no written informed consent was required for the present study, and no further permissions were required. This is in accordance with German National law (Medizinproduktegesetz (MPG) §24).
All authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Förster, I., Brockmann, M., Schildgen, O. et al. Microsatellite instability testing in colorectal cancer using the QiaXcel advanced platform. BMC Cancer 18, 484 (2018) doi:10.1186/s12885-018-4400-z
- Colo-rectal cancer
- QiaXcel advanced