Sentinel lymph node mapping with superparamagnetic iron oxide for melanoma: a pilot study in healthy participants to establish an optimal MRI workflow protocol
BMC Cancer volume 22, Article number: 1062 (2022)
Current pre-operative Sentinel Lymph Node (SLN) mapping using dual tracing is associated with drawbacks (radiation exposure, logistic challenges). Superparamagnetic iron oxide (SPIO) is a non-inferior alternative for SLN mapping in breast cancer patients. Limited research has been performed on SPIO use and pre-operative MRI in melanoma patients to identify SLNs.
Healthy participants underwent MRI-scanning pre- and post SPIO-injection during 20 min. Workflow protocols varied in dosage, massage duration, route of administration and injection sites. The first lymph node showing a susceptibility artefact caused by SPIO accumulation was considered as SLN.
Artefacts were identified in 5/6 participants. Two participants received a 0.5 ml subcutaneous injection and 30-s massage, of which one showed an artefact after one hour. Four participants received a 1.0 ml intracutaneous injection and two-minute massage, leading to artefacts in all participants. All SLNs were observed within five minutes, except after lower limb injection (30 min).
SPIO and pre-operative MRI-scanning seems to be a promising alternative for SLN visualization in melanoma patients. An intracutaneous injection of 1.0 ml SPIO tracer, followed by a two-minute massage seems to be the most effective technique, simplifying the pre-operative pathway. Result will be used in a larger prospective study with melanoma patients.
ClinicalTrials.gov (NCT05054062) – September 9, 2021.
Sentinel lymph node biopsy (SLNB) is an important prognostic procedure in the staging of malignant melanomas and plays a crucial role in postoperative management . The importance of the SLNB has become even more pronounced since the introduction of adjuvant immunotherapy in stage III melanoma . Previous research showed that SLN locations can be unpredictable and therefore difficult to localize without imaging . The current gold standard for SLN mapping is a dual tracing technique, using a Technetium (99mTc) tracer and Blue Dye . This SLN localization technique, however, is associated with several drawbacks . The major challenge is the short half-life of 99mTc, causing logistic challenges in scheduling SLN mapping (lymphoscintigraphy (LS) and single-photon emission computed tomography (SPECT/CT)) prior to the surgery. Additionally, potential radiation exposure for both patients and health care staff require equipment and protocols for preparation and disposal of radioactive isotopes that are not available in all hospitals . Severe allergic reactions to Blue Dye have been described, in addition to the blue skin discoloration at the injection site of Blue Dye [4, 6, 7].
A novel technique that has been studied in breast cancer patients is superparamagnetic iron oxide (SPIO) tracing, using magnetic nanoparticles . In contrast to isotopes, SPIO nanoparticles have an extended life span , providing easier logistics in the preoperative planning. Additionally, SPIO does not carry the risk of radiation exposure to both patients and healthcare staff. Several studies showed that tracing with SPIO is non-inferior to dual tracing with 99mTc and Blue Dye to identify SLNs in breast cancer patients [4, 8, 10,11,12]. The SPIO tracer Magtrace® (Endomagnetics Ltd, London, UK) is the successor of the Sienna + ® tracer (Endomagnetics Ltd, London, UK) and is a liquid, carboxydextran-coated, non-radioactive lymphatic tracer which can be detected with a handheld localization system (Sentimag®, Endomagnetics Ltd, London, UK) . However, pre-operative scans are still required, as sentinel lymph nodes can be located at several and even unpredictable locations in the human body .
SPIO is also expected to be non-inferior to dual tracing in melanoma patients and has previously been analyzed with the use of Sentimag® alone [13, 14]. One small feasibility study published promising preliminary results on pre-operative magnetic resonance imaging (MRI) using SPIO in melanoma patients . Our research team is planning a larger study comparing the diagnostic accuracy of the new technique with SPIO, MRI and the handheld localization system, to the standard dual tracing technique to identify SLNs in melanoma patients. The optimal workflow of the MRI regarding dose of SPIO and route of administration, duration of massage and timing of MRI have however not been defined, with inconclusive results in previous reports [13, 14, 16,17,18]. Therefore, the aim of this study was to develop an optimal SPIO-MRI workflow protocol in healthy participants to localize SLNs.
Materials and methods
In August 2021, healthy participants were enrolled in a small pilot study, performed at Zuyderland Medical Center, the Netherlands. Since this is a pilot study, no sample size calculation was performed. Participants were enrolled until a feasible MRI workflow protocol was defined. The study adopted a step-by-step approach in which the information on the results of the pre-operative MRI obtained in the first participants leads to an optimalization of the workflow protocol of the following participants. Exclusion criteria were known intolerance/hypersensitivity to iron or dextran compounds and the standard MRI exclusion criteria . Participants provided written consent upon participation. This pilot study was approved by the institutional ethics committee (METC Zuyd, Z2021082) and registered at ClinicalTrials.gov (NCT05054062).
SPIO and MRI
MRI was performed with a Magnetom Avanto fit 1.5 T system from Siemens and an 18-channel body coil. Magtrace® (Endomagnetics Ltd, London, UK) was used as SPIO tracer and is a black-brownish sterile aqueous suspension consisting of carboxydextran-coated particles with a particle size of 60 nm. Magtrace® contains circa 28 mg iron per ml and is intended for subcutaneous use [16, 20, 21].
Participants were placed in the scanner in supine position with arms parallel to the body. Pre- and post-SPIO images were obtained using a T1-weighted volumetric interpolated breath-hold examination Dixon sequence (*fl3d2, repetition time / echo time 6.64/4.77 ms; field of view 46.2 cm; Bandwidth 580 Hz; slice thickness 2 mm; imaging time 30 s.). Dependent on participants’ length, injection site and potential SLN regions , body parts were separately scanned pre-SPIO. Subsequently, SPIO was injected sub- or intracutaneously followed by a massage, while the participant was still positioned in the scanner. Post-SPIO images were obtained and merged to whole-body scan images. Images were analyzed by two radiologists using a Picture Archiving and Communication System workstation (Sectra). SPIO uptake from the injection sites and the migration through lymphatic vessels to SLNs was assessed. The first appearing lymph node (LN), showing susceptibility artefacts of SPIO accumulation, was considered to be the SLN. The LN regions were identified.
In order to find the optimal workflow protocol, the injection sites, dosages of Magtrace®, massage duration and route of administration varied among protocols. The first two participants, anterior upper trunk and lower limb, received a subcutaneous injection with 0.5 ml SPIO suspension, followed by a 30 s vigorous massage. First post-injection imaging was performed approximately two minutes after injection. Every other minute, images were produced and repeated until 10 min, then, every two minutes until 20 min. When after 20 min no susceptibility artefacts in LN regions were noted on MRI, scanning was repeated one hour post-contrast. Based on the findings of the first two participants, the protocol for the next four participants was changed with higher dose of 1.0 ml SPIO suspension, with an intracutaneous injection rather than subcutaneous (anterior and posterior trunk, lower arm and lower leg), and with a longer two-minute vigorous massage. Post-injection MR imaging was performed at the same time intervals as in the first two participants. Eighteen days post-injection, images of SPIO distribution to other LNs/regions was evaluated. Detailed information about SPIO location, dosage, massage duration, route of administration and scan area per injection site can be found in Fig. 1 in the results section.
Participants and artefacts
All six participants were healthy and five out of six were male. The mean age of the participants was 37.5 years (range 27—61 years). An artefact at the injection site could be identified on T1-weighted post-contrast images in all participants. An artefact in SLNs was identified in five of six participants. In four participants (1.0 ml SPIO), the lymphatic tract could be identified from injection site up to the LN. Participant characteristics can be found in Table 1.
Time of appearance of LNs
A summary of the time of lymphatic uptake and affected LN regions is shown in Fig. 1. After subcutaneous injection of 0.5 ml SPIO (anterior upper trunk and lower leg), no artefacts were identified in LN regions in the first 20 min. At one hour post-injection, SPIO injected in the upper trunk in participant one drained to the right axillary region. In participant two no artefacts were identified after 0.5 ml SPIO injection in the lower leg. A 1.0 ml SPIO injection administered in the lower arm in participant three drained into the left axillary region. The anterior and posterior trunk of participants four and six drained into both groins and right axillary region, respectively. All artefacts were observable on the MRI within five minutes. At the late scan at 18 days there were other LNs in the same region with susceptibility artefacts, without artefacts in other LN regions. Regarding the lower leg location, no artefacts in LNs were seen with a 0.5 ml SPIO subcutaneous injection in participant two, whereas with the 1.0 ml dose intracutaneously in participant five SPIO uptake was observed after 30 min in the left groin.
Skin discoloration at the injection site was noted in the four participants who received the 1.0 ml intracutaneous injection (Fig. 5). Susceptibility artefacts on MRI persists for a longer period of time.
This pilot study developed a feasible and safe pre-operative SPIO-MRI workflow protocol in healthy subjects. A dose of 1.0 ml SPIO is injected intracutaneously while the patient is positioned in the scanner, followed by a two-minute vigorous massage. MRI scanning commences within five minutes of intracutaneous injection. This SPIO-MRI workflow protocol can be used in subsequent studies comparing this new SLN mapping technique with the current gold standard of 99mTc and Blue Dye in melanoma patients.
Previous studies focused on the “comparison” between SLN identification with SPIO (whether or not in combination with MRI) and dual tracing with isotopes and LS + SPEC/CT. However, workflow protocols varied among studies. Dosage of SPIO varied between 0.5 ml and 2 ml, whether or not diluted with saline up to 5.0 ml, route of administration was either described as subcutaneous or intracutaneous and massage duration ranged between no massage and 20 min. Moreover, MRI timing between pre- and post-injection imaging varied between protocols [12,13,14, 16,17,18].
In a previous study, the dosage did not affect the identification of SLNs, nor the time to first appearance of SLNs . The present study showed that the dosage of SPIO (0.5 ml vs. 1.0 ml) affected the SLN identification. A 0.5 ml subcutaneous injection with SPIO did not lead to SLN identification in one of two participants. However, in the present study a lengthened time of SPIO migration to SLNs in the lower dosage was observed. In the second participant, an artefact was only observed after one hour. In contrast with the latter, 1.0 ml intracutaneous injection showed immediate uptake and SPIO distributed to the SLN within 5 min. Similar to SPIO volume, distance from injection site to SLNs tends to affect the time of first appearance of SLNs. Time of lymphatic uptake of SPIO injected in the participants’ trunk or upper limb differed from time of lymphatic uptake of SPIO injected in the lower limb. This was 5 min versus 30 min, respectively. Migration time of SPIO may therefore differ, depending on the location of the injection site. Findings of time to lymphatic SPIO uptake are consistent with clinical experience in breast cancer patients (uptake within 5 min). Lymphatic drainage in.
the lower limb might be improved by active movement of the leg. This will be investigated further in a planned larger prospective study with melanoma patients.
Previous studies have shown that SPIO is still detectable 30 days post injection. Consistent with our findings, SPIO artefacts were still clearly observable on MRI 18 days post injection [18, 22, 23]. However, no distinction could be made between the SLNs and other LNs on MRI 18 days after SPIO injection. Detection with the handheld magnetometer was not performed after 18 days, as this was not part of the research objective. Based on previous research and current findings, extending the timeframe to seven days (from SPIO injection to SLN harvesting) as described by the Magtrace® information documents [20, 21], seems feasible and beneficial for pre-operative planning and eliminating logistic challenges.
Skin discoloration might be considered a drawback, however, discoloration tends to vanish over time and the injection site is mainly excised as part of WLE in patients with melanoma.
As this is a small feasibility study, the limited number of included participants was insufficient to perform statistical analyses.
This pilot study established a pre-operative SPIO-MRI workflow protocol in healthy volunteers, a protocol that can be used to identify SLNs in patients with melanoma. An intracutaneous injection of 1.0 ml SPIO tracer, followed by a two-minute massage and MRI scanning after 5 min is an effective technique. The value of this technique to identify SLNs in patients with melanoma should be studied first in a prospective feasibility study in patients. This seems to be a promising alternative for SLN visualization in patients with melanoma. The next step is to start a feasibility study in patients with melanoma using this MRI workflow protocol.
Availability of data and materials
All data generated or analyzed during this study is included in this published article.
Stiegel E, Xiong D, Ya J, Funchain P, Isakov R, Gastman B, Vij A. Prognostic value of sentinel lymph node biopsy according to Breslow thickness for cutaneous melanoma. J Am Acad Dermatol. 2018;78(5):942–8.
Keung EZ, Gershenwald JE. The eighth edition American Joint Committee on Cancer (AJCC) melanoma staging system: implications for melanoma treatment and care. Expert Rev Anticancer Ther. 2018;18(8):775–84.
Thompson JF, Uren RF, Shaw HM, McCarthy WH, Quinn MJ, O’Brien CJ, Howman-Giles RB. Location of sentinel lymph nodes in patients with cutaneous melanoma: new insights into lymphatic anatomy. J Am Coll Surg. 1999;189(2):195–204.
Thill M, Kurylcio A, Welter R, van Haasteren V, Grosse B, Berclaz G, Polkowski W, Hauser N. The Central-European SentiMag study: sentinel lymph node biopsy with superparamagnetic iron oxide (SPIO) vs. radioisotope. Breast. 2014;23(2):175–9.
Karakatsanis A, Olofsson H, Stålberg P, Bergkvist L, Abdsaleh S, Wärnberg F. Simplifying Logistics and Avoiding the Unnecessary in Patients With Breast Cancer Undergoing Sentinel Node Biopsy. A Prospective Feasibility Trial of the Preoperative Injection of Super Paramagnetic Iron Oxide Nanoparticles. Scand J Surg. 2018;107(2):130–7.
Shams S, Lippold K, Blohmer JU, Röhle R, Kühn F, Karsten MM: A Pilot Study Evaluating the Effects of Magtrace® for Sentinel Node Biopsy in Breast Cancer Patients Regarding Care Process Optimization, Reimbursement, Surgical Time, and Patient Comfort Compared With Standard Technetium(99). Ann Surg Oncol 2020.
Alvarado MD, Mittendorf EA, Teshome M, Thompson AM, Bold RJ, Gittleman MA, Beitsch PD, Blair SL, Kivilaid K, Harmer QJ, Hunt KK. SentimagIC: A Non-inferiority Trial Comparing Superparamagnetic Iron Oxide Versus Technetium-99m and Blue Dye in the Detection of Axillary Sentinel Nodes in Patients with Early-Stage Breast Cancer. Ann Surg Oncol. 2019;26(11):3510–6.
Mok CW, Tan SM, Zheng Q, Shi L. Network meta-analysis of novel and conventional sentinel lymph node biopsy techniques in breast cancer. BJS Open. 2019;3(4):445–52.
Endomag (2021). Endomag - magtrace - overview. Retrieved on March 16, 2021. [https://www.endomag.com/magtrace/overview/].
Piñero-Madrona A, Torró-Richart JA, de León-Carrillo JM, de Castro-Parga G, Navarro-Cecilia J, Domínguez-Cunchillos F, Román-Santamaría JM, Fuster-Diana C, Pardo-García R. Superparamagnetic iron oxide as a tracer for sentinel node biopsy in breast cancer: A comparative non-inferiority study. European Journal of Surgical Oncology (EJSO). 2015;41(8):991–7.
Ghilli M, Carretta E, Di Filippo F, Battaglia C, Fustaino L, Galanou I, Di Filippo S, Rucci P, Fantini MP, Roncella M. The superparamagnetic iron oxide tracer: a valid alternative in sentinel node biopsy for breast cancer treatment. Eur J Cancer Care. 2017;26(4):e12385.
Pouw JJ, Grootendorst MR, Bezooijen R, Klazen CA, De Bruin WI, Klaase JM, Hall-Craggs MA, Douek M, Ten Haken B. Pre-operative sentinel lymph node localization in breast cancer with superparamagnetic iron oxide MRI: the SentiMAG Multicentre Trial imaging subprotocol. Br J Radiol. 2015;88(1056):20150634.
Piñero-Madrona A, Nicolás-Ruiz F, Rull-Ortuño R, Vidal-Sicart S, Cabañas-Montero J, Rioja-Martín ME, Rodríguez-Fernández R, Gil-Olarte MÁ, González-García B, Sánchez JH. Correlation between ferromagnetic and isotopic tracers for sentinel lymph node detection in cutaneous melanoma: IMINEM study. J Surg Oncol. 2021;123(2):654–9.
Anninga B, White SH, Moncrieff M, Dziewulski P, Geh JLC, Klaase J, Garmo H, Castro F, Pinder S, Pankhurst QA, Hall-Craggs MA, Douek M, MELAMAG MulticentreTrialists Group. Magnetic Technique for Sentinel Lymph Node Biopsy in Melanoma: The MELAMAG Trial. Ann Surg Oncol. 2016;23(6):2070–8.
Katsarelias D, Petrusson H, Leonhardt H, Zaar P, Mattsson J, Bagge RO. Sentinel Node Localisation and Staging with Superparamagnetic Iron Oxide-Enhanced MRI and Sentimag in Patients with Malignant Melanoma. The MAGMEN Feasibility Study. EJSO. 2020;46(2):e131.
Pouw JJ, Ahmed M, Anninga B, Schuurman K, Pinder SE, Van Hemelrijck M, Pankhurst QA, Douek M, Ten Haken B. Comparison of three magnetic nanoparticle tracers for sentinel lymph node biopsy in an in vivo porcine model. Int J Nanomedicine. 2015;10:1235–43.
Douek M, Klaase J, Monypenny I, Kothari A, Zechmeister K, Brown D, Wyld L, Drew P, Garmo H, Agbaje O, Pankhurst Q, Anninga B, Grootendorst M, Ten Haken B, Hall-Craggs MA, Purushotham A, Pinder S. SentiMAG Trialists Group: Sentinel node biopsy using a magnetic tracer versus standard technique: the SentiMAG Multicentre Trial. Ann Surg Oncol. 2014;21(4):1237–45.
Hersi AF, Pistiolis L, Dussan Luberth C, Vikhe-Patil E, Nilsson F, Mohammed I, Olofsson Bagge R, Wärnberg F, Eriksson S, Karakatsanis A. Optimizing Dose and Timing in Magnetic Tracer Techniques for Sentinel Lymph Node Detection in Early Breast Cancers: The Prospective Multicenter SentiDose Trial. Cancers (Basel). 2021;13(4):693. https://doi.org/10.3390/cancers13040693.
Conduct Science. Magnetic Resonance Imaging (MRI) in Research (April 25, 2019). Retrieved on March 24, 2021. [https://conductscience.com/magnetic-resonance-imaging-mri/].
Endomagnetics. Highlights of instructions for use - Magtrace (July, 2018). Retrieved on March 24, 2021. [https://www.accessdata.fda.gov/cdrh_docs/pdf16/P160053D.pdf].
Sysmex Europe, Endomagnetics: Sentimag - Magnetic localisation of sentinel lymph nodes - easy flexible and effective. Retrieved on March 24, 2021. https://www.sysmex.nl/fileadmin/media/f100/Business_Lines/Brochure/Magtrace_brochure_EN_01.pdf.
Karakatsanis A, Hersi AF, Pistiolis L, Olofsson Bagge R, Lykoudis PM, Eriksson S, Wärnberg F. SentiNot Trialists Group: Effect of preoperative injection of superparamagnetic iron oxide particles on rates of sentinel lymph node dissection in women undergoing surgery for ductal carcinoma in situ (SentiNot study). Br J Surg. 2019;106(6):720–8.
Wärnberg F, Stigberg E, Obondo C, Olofsson H, Abdsaleh S, Wärnberg M, Karakatsanis A. Long-Term Outcome After Retro-Areolar Versus Peri-Tumoral Injection of Superparamagnetic Iron Oxide Nanoparticles (SPIO) for Sentinel Lymph Node Detection in Breast Cancer Surgery. Ann Surg Oncol. 2019;26(5):1247–53.
Ethics approval and consent to participate
Approved by METC Zuyderland-Zuyd (METC Zuyd, Z2021082). The study was conducted in accordance with the declaration of Helsinki. Informed consent was obtained from all subjects/participants and/or legal guardian(s).
Consent for publication
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Aldenhoven, L., Frotscher, C., Körver-Steeman, R. et al. Sentinel lymph node mapping with superparamagnetic iron oxide for melanoma: a pilot study in healthy participants to establish an optimal MRI workflow protocol. BMC Cancer 22, 1062 (2022). https://doi.org/10.1186/s12885-022-10146-w
- Magnetic iron oxide nanoparticles
- Sentinel lymph node biopsy
- Surgical oncology