- Research article
- Open Access
Use of the QIAGEN GeneReader NGS system for detection of KRAS mutations, validated by the QIAGEN Therascreen PCR kit and alternative NGS platform
BMC Cancer volume 17, Article number: 358 (2017)
The detection of somatic mutations in primary tumors is critical for the understanding of cancer evolution and targeting therapy. Multiple technologies have been developed to enable the detection of such mutations. Next generation sequencing (NGS) is a new platform that is gradually becoming the technology of choice for genotyping cancer samples, owing to its ability to simultaneously interrogate many genomic loci at massively high efficiency and increasingly lower cost. However, multiple barriers still exist for its broader adoption in clinical research practice, such as fragmented workflow and complex bioinformatics analysis and interpretation.
We performed validation of the QIAGEN GeneReader NGS System using the QIAact Actionable Insights Tumor Panel, focusing on clinically meaningful mutations by using DNA extracted from formalin-fixed paraffin-embedded (FFPE) colorectal tissue with known KRAS mutations. The performance of the GeneReader was evaluated and compared to data generated from alternative technologies (PCR and pyrosequencing) as well as an alternative NGS platform. The results were further confirmed with Sanger sequencing.
The data generated from the GeneReader achieved 100% concordance with reference technologies. Furthermore, the GeneReader workflow provides a truly integrated workflow, eliminating artifacts resulting from routine sample preparation; and providing up-to-date interpretation of test results.
The GeneReader NGS system offers an effective and efficient method to identify somatic (KRAS) cancer mutations.
Somatic mutations in the KRAS oncogene are common in human cancers. They are found in 70-90% of pancreatic cancers [1, 2], 30-50% of colorectal cancers [3,4,5] and 10-30% of Non-Small Cell Lung Cancers (NSCLC) [6,7,8]. Several methods have been developed for the detection of KRAS mutations, each with specific advantages and limitations [5, 9, 10].
Sanger sequencing has been the ‘gold standard’ for mutation analysis in cancer detection since the 1970s . However, limited by its low sensitivity (10-20% mutant allele frequency (MAF)) and low throughput , Sanger sequencing is no longer sufficient for the needs of today’s cancer molecular diagnostics.
The therascreen KRAS RGQ PCR kit is a real-time qPCR-based assay used to detect the most common KRAS mutations including those in codons 12 and 13. It has greatly improved sensitivity over Sanger sequencing, and has been approved by the Food and Drug Administration (FDA)  for colorectal cancer patient stratification. Pyrosequencing also offers an attractive alternative to Sanger due to its fast turnaround time (TAT) and lower sensitivity threshold, even in tissues with low tumor cell content .
Next-generation sequencing (NGS) differs radically from the above mentioned methods. Coupled with amplicon-based targeting technology, NGS has the capability to simultaneously sequence in a massively parallel way multiple genetic loci with minimal amounts of nucleic acid input and limited time and expense [12,13,14,15]. This technology has revolutionized the speed of genetic and genomic discovery, and advanced our understanding of molecular mechanisms of diseases. In recent years, NGS has played an important role in advancing personalized healthcare and precision medicine by enabling the identification of mutations associated with therapeutic response or resistance. As more clinically significant genetic biomarkers and targeted therapies become available, the profiling of such genetic variations is becoming increasingly more critical. Several NGS platforms are already commercially available for sequencing and identification of genetic alterations associated with diseases, such as point mutations, deletions, insertions and copy number variants . However, QIAGEN’s GeneReader System presented here includes all upstream sample processing steps starting from nucleic acid extraction, together with an integrated downstream bioinformatics solution that enables a direct access to real-time updates from the rapidly evolving literature, and clinical knowledge and evidence.
To this end, we recently evaluated the QIAGEN GeneReader System workflow from DNA extraction and purification from FFPE tissue samples, to library preparation, sequencing and data analysis and interpretation. Herein we show that the GeneReader presents a unified workflow that provides accurate results and a simple solution for any laboratory to use in clinical research.
Sample and DNA isolation
FFPE Tumor material from colorectal cancer tumors (Origene Technologies, MD, USA and Asterand Biosciences, MI, USA) was used to prepare 56 DNA samples with known KRAS mutation status, previously determined using therascreen assay (Pyrosequencing and PCR) and Sanger sequencing according to methods further described below. Tissue sections of 10 μm in thickness, ranging from 3 to 20 years of age were used for DNA extraction utilizing either: i) the QIAamp DNA FFPE Tissue Kit (QIAGEN, Hilden, Germany) or ii) the GeneRead FFPE DNA Kit (QIAGEN, Hilden, Germany) according to manufacturer’s instructions. DNA concentration was determined using the Nanodrop System (Thermo Fisher Scientific, MA, USA) and Qubit dsDNA HS assay (Life Technologies, Gaithersburg, USA). The DNA was assessed using the GeneRead DNA QuantiMIZE System (QIAGEN, Hilden, Germany) which utilizes a qPCR-based approach to determine the quality of sample DNA prior to NGS. Furthermore, both NA12878 (Coriell Institute for Medical Research) (for which the Genome in the Bottle (GIAB) consortium has published a set of high confident variants ) and AcroMetrix (Thermo Fisher Scientific, MA, USA) samples were used as a gold standard set of variant calls.
GeneReader sample preparation and sequencing run
In total, 40 ng of DNA measured by Qubit (Thermo Fisher Scientific, MA, USA) was used as template to generate libraries for sequencing. Libraries were prepared using the QIAGEN Library Kit v2.0 and the GeneRead QIAact Actionable Insight Tumor Panel (QIAGEN, Hilden, Germany), which amplifies 330 amplicons covering 16.7 kb, containing 773 unique variant positions in 12 genes (KRAS, NRAS, KIT, BRAF, PDGFRA, ALK, EGFR, ERBB2, PIK3CA, ERBB3, ESR1 and RAF1). All steps of library preparation were performed according to the manufacturer’s protocol. The libraries were then quantified using a Qubit dsDNA HS Assay Kit (Life Technologies, MA, USA) and QIAxcel (QIAGEN, Hilden, Germany). Ten individual libraries were pooled prior to emulsion PCR and bead enrichment steps that were carried out using an automated protocol on the GeneRead QIAcube (QIAGEN, Hilden, Germany) using the GeneRead Clonal Amp Q Kit (QIAGEN, Hilden, Germany), according to the manufacturer’s protocol. Following bead enrichment, the pooled libraries were sequenced using the GeneReader platform (QIAGEN, Hilden, Germany).
GeneReader data processing
QIAGEN Clinical Insight (QCI™) Analyze software (QIAGEN, Hilden, Germany) was used to QC, align the read data to the hg19 reference genome sequence, call sequence variants, and generate an interactive report for visualization of the sequencing results, as well as a summary of the data. QCI Analyze software reports a set of high- and low-confidence variants based on the coverage of variant positions. Users have an option to analytically confirm if a variant listed should be valid or invalid before uploading to QCI Interpret software for the clinical interpretation. For each sample the report was used to assess the quality of the overall sequencing run and to identify/call the individual variants. After review, variants confirmed as analytically valid were uploaded to QCI Interpret for creation of a report for each sample based on detected variants and curated content, with a summary of findings and direct links to evidence sources.
The Actionable Insight Tumor Panel (QIAGEN, Hilden, Germany) was used for a MiSeq (Illumina, CA, USA) sequencing run. The Kapa “with bead” PCR free protocol (KAPABiosystems, MA, USA) was used in further Illumina library preparation steps. Samples were then paired-end sequenced on a MiSeq instrument (Illumina, CA, USA) according to Illumina guidelines. The resulting reads were mapped to the hg19 reference genome sequence using BWA mem software followed by GATK (best practices) to recalibrate base quality scores. Variants were called using MuTect. Variants were then filtered using GATK (best practice) and annotated using SnpEff. Variants at hotspot positions were selected using GATK.
Pyrosequencing and Sanger analyses
The sample DNA obtained with the QIAamp FFPE DNA Kit (QIAGEN, Hilden, Germany) was subjected to Pyrosequencing analysis and Sanger sequencing. For Pyrosequencing the samples were analyzed using the therascreen RAS Extension Pyro Kit (QIAGEN, Hilden, Germany) which covers mutations in KRAS codons 59, 61, 117 and 146 as well as NRAS codons 59, 117 and 146. Samples with mutations in KRAS or NRAS codons 12 and 13 were further analyzed with the therascreen KRAS or NRAS Pyro Kit (QIAGEN, Hilden, Germany) according to manufacturer’s instructions. In addition, samples that failed the initial PyroMark KRAS analysis were subjected to a second round of analysis. Samples with an initial “check” status, or with an indicated mutation signal of LOD + 3% (“Potential low level mutation”) were subjected to a second round of analysis performed in duplicate. Sanger sequencing was performed using Big Dye Terminator Technology and an ABI 3730xl sequencer (Thermo Fisher Scientific, MA, USA). Mutations were detected by analyzing the sequence trace files and the quantity of a base at a certain position was calculated from the area under the curve (AUC) at the mutation specific position in the electropherogram.
The therascreen KRAS RGQ PCR Kit (QIAGEN, Hilden, UK) is an allele-specific PCR-based technology with specific primers for the seven most common KRAS codon 12 and 13 mutations. The assay screens for the following mutations: 12 GCT (Ala), 12 GAT (Asp), 12 CGT (Arg), 12 TGT (Cys), 12 AGT (Ser), 12 GTT (Val), and 13 GAC (Asp). Mutation analysis was performed according to manufacturer’s instructions, using the RotorGene real-time PCR instrument (QIAGEN, Hilden, UK). Analysis of results was performed following the recommendations in the manual, e.g. samples with a control assay with a cycle threshold (Ct) of 35 or higher were deemed invalid and excluded from the analysis. Samples were called mutation positive based on the delta Ct values reported in the handbook. Values over 40 cycles were scored as negative (wild-type).
Evaluation of DNA quality by QuantiMIZE
FFPE samples with ages ranging from 3 to 20 years were used for this study. The quality of the extracted DNA was measured by the GeneRead DNA QuantiMIZE QC assay (QIAGEN, Hilden, UK). Thirteen out of 56 samples failed quality checks and were excluded from further analysis (Additional file 1: Table S1). For the remaining 43 samples, 3 to 9 PCR cycles were added (depending on the QuantiMIZE quality scores) to compensate for differences in DNA quality during enrichment PCR. The additional cycles ensured that poor quality (highly fragmented) DNA samples yielded enough material for downstream library preparation. The quality of DNA purified from formalin fixated tissue decreases over the sample storage period time [18,19,20], but also depends on how tissues were treated, handled and processed before and during sample fixation [19, 21, 22].
GeneReader sequencing performance
The QIAact Actionable Insights Tumor Panel (QIAGEN, Hilden, UK) contains 773 unique variant positions in 12 genes (Table 1). An analysis of the reads mapped to the reference showed coverage levels that met the industry-standard 5% sensitivity criteria, even with aged FFPE samples. A 200× minimum read coverage cutoff was used for calling a variant at any position in the panel. For the 43 FFPE samples analyzed, an average amplicon coverage of 97.2% was observed, and an average variant insight coverage (hotspot coverage) of 99.8% was observed at read depths ≥200× (Table 1). For NA12878 samples, an average amplicon coverage of 98.5% was observed and an average variant insight coverage of 99.9% was observed at read depths of ≥200× (Table 1). No false negatives (FN; where an expected variant was not detected) were observed.
Performance comparison between the QIAamp and GeneRead DNA FFPE kits for DNA purification using the GeneReader
Two DNA purification kits were used to isolate DNA from FFPE samples. Table 2 demonstrates the superior performance of the GeneRead DNA FFPE Kit (QIAGEN, Hilden, UK) over the QIAamp DNA FFPE Tissue Kit (QIAGEN, Hilden, UK) in terms of true positives at lower variant calling sensitivity. Fourteen true positive KRAS variants were detected using an allele fraction cut-off of >5% for DNA isolated by GeneRead DNA FFPE Kit (QIAGEN, Hilden, UK). For the QIAamp DNA FFPE Tissue Kit (QIAGEN, Hilden, UK), 15 KRAS variants were detected using an allele fraction cut-off of >5%. Of the 15 KRAS variants detected, 14 were true positive variants and 1 was a false positive (Table 2) as confirmed by several independent methods. Decreasing the allele fraction cutoff to >2.5% resulted in identification of the same 14 KRAS true positive samples for GeneRead DNA FFPE Kit (QIAGEN, Hilden, UK) extractions. However, for QIAamp DNA FFPE Tissue Kit (QIAGEN, Hilden, UK) extracted samples at >2.5% allele fraction cut-off, 11 additional false positive KRAS mutations (25 variants in total) were detected. The additional mutations were mostly C to T transitions. It is known that FFPE fixation deaminates certain bases, most prominently cytosine deamination to uracil [23,24,25]. The GeneRead DNA FFPE Kit (QIAGEN, Hilden, UK) contains an integrated uracil DNA glycosylase (UDG) step which removes uracil from the DNA before the final purification step, yielding high-quality DNA with minimal artifacts.
Confirmation of variants by MiSeq, pyrosequencing and therascreen qPCR assays
The GeneReader NGS System variant calls demonstrated 100% agreement with KRAS mutation status previously determined by either pyrosequencing or therascreen qPCR (Table 3). Of the 43 samples, 14 tested positive for KRAS variants and 29 samples were confirmed as wild type. The 5% allelic fraction cut-off was used to call KRAS variants for codons 12, 13, 59, 61, 117 and 146. The true positive variants observed by the GeneReader NGS System share a 100% concordance with MiSeq-Illumina (Table 4).
The use of the NA12878 control (Fig. 1, Additional file 2: Table S2) and AcroMetrix (Fig. 1, Additional file 3: Tables S3) reference standard materials demonstrated the good performance of the GeneReader platform on high frequency and low frequency variants, respectively. NA12878 has been used extensively as a reference standard material for verifying NGS platforms  and acts as a useful control in establishing background error. Besides its use as a GeneReader platform performance standard, AcroMetrix has also been used previously as a control for variant calls .
A major advantage of NGS over traditional mutation detection methods is the ability to sequence multiple genes and variants simultaneously. Other advantages include minimal DNA input, faster turnaround time; lower overall cost and higher throughput and sensitivity compared to traditional methods [12, 27,28,29]. NGS has revolutionized the speed of genetic and genomic discovery, and advanced our understanding of the molecular mechanisms of disease and potential treatment options. However, several major hurdles remain and still prevent NGS from being broadly adopted in clinical practice. This is especially true for laboratories that are new to this technology, and may lack the in-house expertise required for processing complex bioinformatics data and interpretation of results. Such expertise is crucial to construct a bioinformatics pipeline and to evaluate the software and generate quality reports. The QIAGEN GeneReader NGS System allows users to perform experiments from sample to insight, tissue sample to decipherable report based on the interpretation of sequence variants detected.
The QIAGEN GeneReader NGS workflow utilizes ‘QCI Analyze’ and ‘QCI Interpret’ for bioinformatics analysis and reporting of variants, including read mapping, variant calling and interpretation of results. It provides visualization of the alignment of sequencing results (Fig. 2) as well as a summary of the data. Quality assessment is also supported, both at the overall sequencing run level and for the analytic validity of individual variants to reduce false positive and negative results. Using the data visualization tools within QCI Analyze, it is possible to determine the quality of the results and assess any variants of interest. Further analysis of variants using QCI Interpret provides access to the curated information contained within the QIAGEN Knowledge Base enabling a deeper analysis and interpretation of results for each sample (Fig. 3). With all relevant information, a report can be created with a summary of findings and direct links to evidence sources. At the single variant level the QCI software is able to identify an individual variant as an actionable cancer mutation, and provides links to current clinical research insights, e.g. the KRAS G12D somatic variant it is established to confer resistance to the colorectal cancer drugs cetuximab and panitumumab, based on evidence curated from their FDA drug labels and clinical practice guidelines. Within QCI-Interpret information on active clinical trials recruiting colorectal cancer patients with particular mutations are provided with drug, nearest location, and trial phase information.
The relationship between FFPE DNA quality and sequencing accuracy is a critical point for any sequencing analysis. The GeneReader workflow starts with the GeneRead FFPE DNA Kit for DNA extraction and is specifically designed to reduce artifacts known to commonly occur in FFPE treated samples. As seen in Table 2, by using FFPE samples aged from 3 to 20 years, the GeneRead FFPE DNA Kit successfully reduced the number of low frequency false positive variants detected. These low frequency false positive variants are likely caused by cytosine deamination and other fixation associated artifacts. Similar phenomena were observed by Bourgon , where pretreatment of FFPE samples with uracil DNA glycosylase (UDG) resulted in a dramatic reduction of false positives, with overall reductions of 77% for C > T and 94% for G > A changes, respectively. Biochemical removal of deaminated DNA eliminates deamination-associated false positive results; however, for samples with very low quality DNA such as highly fragmented FFPE treated samples, UDG-treated may constitute an issue, as the treatment introduces possible further strand breaks leading to even higher fragmentation and lower availability of intact template strands. Therefore, using the QuantiMIZE assay to identify those samples suitable for sequencing, based on an assessment of original intact and amplifiable templates, before starting an experiment is a critical point for an amplification based NGS technology. Previous reports observed that samples with lower amounts of amplifiable DNA are more likely to give a markedly increased number of false positive results [30, 31].
In summary, this study confirms that the GeneReader NGS System performs consistently and accurately in the identification of somatic mutations from FFPE samples, with results confirmed by both alternative technologies as well as an alternative NGS platform. With a full end-to-end solution with integrated sample preparation and bioinformatics interpretation, the GeneReader NGS System is suitable for any laboratory interested in cancer clinical research.
Food and Drug Administration
Next generation sequencing
Non-small cell lung cancer
Polymerase chain reaction
QIAGEN clinical insight
Di Marco M, Astolfi A, Grassi E, Vecchiarelli S, Macchini M, Indio V, Casadei R, Ricci C, D’Ambra M, Taffurelli G, Serra C, Ercolani G, Santini D, D’Errico A, Pinna AD, Minni F, Durante S, Martella LR, Biasco G. Characterization of pancreatic ductal adenocarcinoma using whole transcriptome sequencing and copy number analysis by single-nucleotide polymorphism array. Mol Med Rep. 2015;12:7479–84.
Huang J, Löhr JM, Nilsson M, Segersvärd R, Matsson H, Verbeke C, Heuchel R, Kere J, Iafrate AJ, Zheng Z, Ye W. Variant profiling of candidate genes in pancreatic Ductal Adenocarcinoma. Clin Chem. 2015;61:1408–16.
Gao J, Wu H, Wang L, Zhang H, Duan H, Lu J, Liang Z. Validation of targeted next-generation sequencing for RAS mutation detection in FFPE colorectal cancer tissues: comparison with Sanger sequencing and ARMS-scorpion real-time PCR. BMJ Open. 2016;6:e009532.
Sakai K, Yoneshige A, Ito A, Ueda Y, Kondo S, Nobumasa H, Fujita Y, Togashi Y, Terashima M, De Velasco MA, Tomida S, Nishio K. Performance of a novel KRAS mutation assay for formalin-fixed paraffin embedded tissues of colorectal cancer. Spring. 2015;4:7.
Sundström M, Edlund K, Lindell M, Glimelius B, Birgisson H, Micke P, Botling J. KRAS analysis in colorectal carcinoma: analytical aspects of pyrosequencing and allele-specific PCR in clinical practice. BMC Cancer. 2010;10:660.
Casadio C, Guarize J, Donghi S, Di Tonno C, Fumagalli C, Vacirca D, Dell’Orto P, De Marinis F, Spaggiari L, Viale G, Barberis M. Molecular testing for targeted therapy in advanced non-small cell lung cancer: suitability of Endobronchial ultrasound Transbronchial needle aspiration. Am J Clin Pathol. 2015;144:629–34.
Papadopoulou E, Tsoulos N, Tsirigoti A, Apessos A, Agiannitopoulos K, Metaxa-Mariatou V, Zarogoulidis K, Zarogoulidis P, Kasarakis D, Kakolyris S, Dahabreh J, Vlastos F, Zoublios C, Rapti A, Papageorgiou NG, Veldekis D, Gaga M, Aravantinos G, Karavasilis V, Karagiannidis N, Nasioulas G. Determination of EGFR and KRAS mutational status in Greek non-small-cell lung cancer patients. Oncol Lett. 2015;10:2176–84.
Lièvre A, Bachet JB, Boige V, Cayre A, Le Corre D, Buc E, Ychou M, Bouché O, Landi B, Louvet C, André T, Bibeau F, Diebold MD, Rougier P, Ducreux M, Tomasic G, Emile JF, Penault-Llorca F, Laurent-Puig P. KRAS mutations as an independent prognostic factor in patients with advanced colorectal cancer treated with cetuximab. J Clin Oncol. 2008;26:374–9.
Rodriguez R. Biomarker testing for treatment of metastatic colorectal cancer: role of the pathologist in community practice. J Community Support Oncol. 2014;12:27–32.
Martinez DA, Nelson MA. The next generation becomes the now generation. PLoS Genet. 2010;6:e1000906.
Sanger F, Nicklen S, Coulson AR. DNA sequencing with chain-terminating inhibitors. Proc Natl Acad Sci U S A. 1997;74:5463–7.
D’Haene N, Le Mercier M, De Nève N, Blanchard O, Delaunoy M, El Housni H, Dessars B, Heimann P, Remmelink M, Demetter P, Tejpar S, Salmon I. Clinical validation of targeted next generation sequencing for Colon and Lung cancers. PLoS One. 2015;10:e0138245.
Le Mercier M, D’Haene N, De Nève N, Blanchard O, Degand C, Rorive S, Salmon I. Next-generation sequencing improves the diagnosis of thyroid FNA specimens with indeterminate cytology. Histopathology. 2015;66:215–24.
Beadling C, Neff TL, Heinrich MC, Rhodes K, Thornton M, Leamon J, Andersen M, Corless CL. Combining highly multiplexed PCR with semiconductor-based sequencing for rapid cancer genotyping. J MolDiagn. 2013;15:171–6.
Metzger GJ, Dankbar SC, Henriksen J, Rizzardi AE, Rosener NK, Schmechel SC. Development of multigene expression signature maps at the protein level from digitized immunohistochemistry slides. PLoS One. 2012;7:e33520.
Metzker ML. Sequencing technologies - the next generation. Nat Rev Genet. 2010;11:31–46.
Zook JM, Chapman B, Wang J, Mittelman D, Hofmann O, Hide W, Salit M. Integrating human sequence data sets provides a resource of benchmark SNP and indel genotype calls. Nat Biotechnol. 2014;32:246–51.
Frickmann H, Tenner-Racz K, Eggert P, Schwarz NG, Poppert S, Tannich E, Hagen RM. Influence of parasite density and sample storage time on the reliability of Entamoeba Histolytica-specific PCR from formalin-fixed and paraffin-embedded tissues. Diagn Mol Pathol. 2013;22:236–44.
Obersteller S, Neubauer H, Hagen RM, Frickmann H. Comparison of five commercial nucleic acid extraction kits for the PCR-based detection of Burkholderia Pseudomallei DNA in formalin-fixed, paraffin-embedded tissues. Eur J Microbiol Immunol (Bp). 2016;6:244–52.
Kokkat TJ, Patel MS, McGarvey D, VA LV, Baloch ZW. Archived formalin-fixed paraffin-embedded (FFPE) blocks: a valuable underexploited resource for extraction of DNA, RNA, and protein. Biopreserv Biobank. 2013;11:101–6.
Chung JY, Braunschweig T, Williams R, Guerrero N, Hoffmann KM, Kwon M, Song YK, Libutti SK, Hewitt SM. Factors in tissue handling and processing that impact RNA obtained from formalin-fixed, paraffin-embedded tissue. J Histochem Cytochem. 2008;56:1033–42.
Frickmann H, Loderstaedt U, Racz P, Tenner-Racz K, Eggert P, Haeupler A, Bialek R, Hagen RM. Detection of tropical fungi in formalin-fixed, paraffin-embedded tissue: still an indication for microscopy in times of sequence-based diagnosis? Biomed Res Int. 2015;2015:938721–32.
Bourgon R, Lu S, Yan Y, Lackner MR, Wang W, Weigman V, Wang D, Guan Y, Ryner L, Koeppen H, Patel R, Hampton GM, Amler LC, Wang Y. High-throughput detection of clinically relevant mutations in archived tumor samples by multiplexed PCR and next-generation sequencing. Clin Cancer Res. 2014;20:2080–91.
Hosein AN, Song S, McCart Reed AE, Jayanthan J, Reid LE, Kutasovic JR, Cummings MC, Waddell N, Lakhani SR, Chenevix-Trench G, Simpson PT. Evaluating the repair of DNA derived from formalin-fixed paraffin-embedded tissues prior to genomic profiling by SNP-CGH analysis. Lab Investig. 2013;93:701–10.
Do H, Dobrovic A. Dramatic reduction of sequence artefacts from DNA isolated from formalin-fixed cancer biopsies by treatment with uracil- DNA glycosylase. Oncotarget. 2012;3:546–58.
Masucci GV, Cesano A, Hawtin R, Janetzki S, Zhang J, Kirsch I, Dobbin KK, Alvarez J, Robbins PB, Selvan SR, Streicher HZ, Butterfield LH, Thurin M. Validation of biomarkers to predict response to immunotherapy in cancer: volume I - pre-analytical and analytical validation. J Immunother Cancer. 2016;4:76–101.
Castéra L, Krieger S, Rousselin A, Legros A, Baumann JJ, Bruet O, Brault B, Fouillet R, Goardon N, Letac O, Baert-Desurmont S, Tinat J, Bera O, Dugast C, Berthet P, Polycarpe F, Layet V, Hardouin A, Frébourg T, Vaur D. Next-generation sequencing for the diagnosis of hereditary breast and ovarian cancer using genomic capture targeting multiple candidate genes. Eur J Hum Genet. 2014;22:1305–13.
McCourt CM, McArt DG, Mills K, Catherwood MA, Maxwell P, Waugh DJ, Hamilton P, O’Sullivan JM, Salto-Tellez M. Validation of next generation sequencing technologies in comparison to current diagnostic gold standards for BRAF, EGFR and KRAS mutational analysis. PLoS One. 2013;8:e69604.
Tuononen K, Mäki-Nevala S, Sarhadi VK, Wirtanen A, Rönty M, Salmenkivi K, Andrews JM, Telaranta-Keerie AI, Hannula S, Lagström S, Ellonen P, Knuuttila A, Knuutila S. Comparison of targeted next-generation sequencing (NGS) and real-time PCR in the detection of EGFR, KRAS, and BRAF mutations on formalin-fixed, paraffin-embedded tumor material of non-small cell lung carcinoma-superiority of NGS. Genes Chromosom Cancer. 2013;52:503–11.
Betge J, Kerr G, Miersch T, Leible S, Erdmann G, Galata CL, Zhan T, Gaiser T, Post S, Ebert MP, Horisberger K, Boutros M. Amplicon sequencing of colorectal cancer: variant calling in frozen and formalin-fixed samples. PLoS One. 2015;10:e0127146.
Sah S, Chen L, Houghton J, Kemppainen J, Marko AC, Zeigler R, Latham GJ. Functional DNA quantification guides accurate next-generation sequencing mutation detection in formalin-fixed, paraffin-embedded tumor biopsies. Genome Med. 2013;5:77–89.
We greatly appreciate Drs. Scott Steelman, Kathleen Steinmann and Robert Lintner from Broad Technology Labs (Broad Institute of MIT and Harvard, 75 Ames Street (Rm 8021), Cambridge, MA 02141) for their support in samples processing, GeneReader testing, and manuscript revisions. We thanks to Drs. Vikas Gupta, Naomi Thompson Kiran Divakar and Dietrich Lueerssen from QIAGEN for the input of the manuscript.
This work was supported by QIAGEN.
The sequencing chemistry used in this manuscript is currently available outside the US. The GeneReader NGS System is for research use only. An upgraded and different sequencing chemistry has meanwhile been made available in the US since April 2017 and will become available worldwide later in 2017. For the release of the new sequencing chemistry in the US in April, we have shown equivalency to the sequencing chemistry used in the manuscript.
Availability of data and materials
The analyzed data sets generated during the study are available from the corresponding author on reasonable request.
AD, AMH, SS, AS, EL, SH, TR designed the study. AD, AMH, SS, AS, DR, MN, TP, DA, AR, NP, DS, TR performed experiments and analyzed data. AD, YK and TR wrote the manuscript. AK, DS, DZ, KK assisted in preparing the manuscript. All authors read and approved the final manuscript.
At the time of the work was being done the authors where employees of QIAGEN, and QIAGEN funded the research and the publication costs. We declare that our current or previous employment with QIAGEN did not influence our interpretation of data or presentation of information.
Consent for publication
Ethics approval and consent to participate
FFPE Tumor material from colorectal cancer tumors (Origene Technologies, MD, USA and Asterand Biosciences, MI, USA). We refer to Origene’s and Asterand’s quality system as well as ethics and compliance processes with informed consent for commercial clinical samples. We expect Origene and Asterand to follow industry standard ethics approval and consent processes.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
The QC results of the extracted DNA samples were measured using GeneRead DNA QuantiMIZE. (DOCX 28 kb)
List of NA12878 Gold Standard Variants from 18 samples sequenced by GeneReader. (DOCX 27 kb)
List of AcroMetrix™ Oncology Hotspot Gold Standard Variants from 18 samples sequenced by GeneReader (DOCX 28 kb) (DOCX 27 kb)
About this article
Cite this article
Darwanto, A., Hein, A., Strauss, S. et al. Use of the QIAGEN GeneReader NGS system for detection of KRAS mutations, validated by the QIAGEN Therascreen PCR kit and alternative NGS platform. BMC Cancer 17, 358 (2017). https://doi.org/10.1186/s12885-017-3328-z