KRAS analysis in colorectal carcinoma: Analytical aspects of Pyrosequencing and allele-specific PCR in clinical practice
© Sundström et al; licensee BioMed Central Ltd. 2010
Received: 13 April 2010
Accepted: 1 December 2010
Published: 1 December 2010
Epidermal growth factor receptor inhibitor therapy is now approved for treatment of metastatic colorectal carcinomas (CRC) in patients with tumors lacking KRAS mutations. Several procedures to detect KRAS mutations have been developed. However, the analytical sensitivity and specificity of these assays on routine clinical samples are not yet fully characterised.
The practical aspects and clinical applicability of a KRAS-assay based on Pyrosequencing were evaluated in a series of 314 consecutive CRC cases submitted for diagnostic KRAS analysis. The performance of Pyrosequencing compared to allele-specific, real-time PCR was then explored by a direct comparison of CE-IVD-marked versions of Pyrosequencing and TheraScreen (DxS) KRAS assays for a consecutive subset (n = 100) of the 314 clinical CRC samples.
Using Pyrosequencing, 39% of the 314 CRC samples were found KRAS-mutated and several of the mutations (8%) were located in codon 61. To explore the analytical sensitivity of the Pyrosequencing assay, mutated patient DNA was serially diluted with wild-type patient DNA. Dilutions corresponding to 1.25-2.5% tumor cells still revealed detectable mutation signals. In clinical practice, our algorithm for KRAS analysis includes a reanalysis of samples with low tumor cell content (< 10%, n = 56) using an independent assay (allele-specific PCR, DxS). All mutations identified by Pyrosequencing were then confirmed and, in addition, one more mutated sample was identified in this subset of 56 samples. Finally, a direct comparison of the two technologies was done by re-analysis of a subset (n = 100) of the clinical samples using CE-IVD-marked versions of Pyrosequencing and TheraScreen KRAS assays in a single blinded fashion. The number of samples for which the KRAS codon 12/13 mutation status could be defined using the Pyrosequencing or the TheraScreen assay was 94 and 91, respectively, and both assays detected the same number of codon 12 and 13 mutations.
KRAS mutation detection using Pyrosequencing was evaluated on a consecutive set of clinical CRC samples. Pyrosequencing provided sufficient analytical sensitivity and specificity to assess the mutation status in routine formalin-fixed CRC samples, even in tissues with a low tumor cell content.
Targeting of epidermal growth factor receptor (EGFR) with the monoclonal antibodies cetuximab or panitumumab prevents activation of downstream signalling molecules. Thereby cellular events such as proliferation, migration and survival are affected. Anti-EGFR therapy is recommended for patients with refractory metastatic colorectal cancer (mCRC) and is currently evaluated in clinical studies as a first line therapy in mCRC. However, activating somatic point mutations in Kirsten RAS (KRAS) are strongly associated with resistance to anti-EGFR therapy and are present in approximately 40% of colorectal tumors [1–12]. Therefore, treatment is only approved for patients harbouring a tumor with a wild-type (wt) KRAS gene. Consequently, robust, reliable, and sensitive methods for mutation analysis are required to stratify patients eligible for anti-EGFR therapy. Specific mutations in codon 12, 13 or 61 of the KRAS gene convert the gene into an active oncogene . Mutations in codon 12 or 13 are the most frequent alterations in KRAS and represent more than 90% of all mutations. Analyses of KRAS in CRC clinical trials have therefore focused on these codons when relating KRAS mutational status to objective response or survival during EGFR inhibitor therapy. Despite being described as an activating KRAS mutation in vitro, the frequency of codon 61 mutations in human tumors is generally reported as low [2, 14–18], and the clinical impact of these mutations is still under discussion [19, 20].
The Pyrosequencing technology  has an analytical sensitivity for detection of mutations that is superior to that of Sanger (dideoxy) sequencing. Several in-house Pyrosequencing assays for detection of KRAS mutations in codon 12, 13 and 61 have been developed [12, 14, 22, 23]. However, in a clinical setting there are several challenges when performing mutation analysis on DNA from routinely formalin-fixed, paraffin-embedded (FFPE) tissue samples. These include suboptimal quality of DNA due to formalin fixation, low tumor cell content in tumor tissues with abundant inflammatory cells, and insufficient starting material, e.g. minimal biopsy fragments. The practical aspects and efficiency of the CE-IVD-marked Pyrosequencing kit for analysis of KRAS mutations have until now not been evaluated in a clinical setting.
The TheraScreen kit (DxS Ltd, Manchester, UK) is a well-established CE-IVD-marked kit for diagnostic analysis of KRAS mutational status . The DxS technology combines allele-specific amplification with real-time PCR for analysis of seven mutations in codons 12 and 13. With high quality DNA this method has the potential to detect ≤ 1% mutant alleles in DNA from a tumor tissue. However, this level of sensitivity is difficult to obtain in most diagnostic FFPE samples . The DxS technology was used in many of the retrospective studies that evaluated the response to anti-EGFR therapy in relation to KRAS mutation status [4, 26, 27].
The aims of the present study were: (i) to describe the KRAS mutation spectrum in a consecutive series of CRC specimens (n = 314) referred to our laboratory for KRAS mutation analysis during the last two years; (ii) to evaluate the performance of Pyrosequencing compared to allele-specific PCR (DxS) on the samples (n = 56) that originated from tissues with a low tumor cell count; and (iii) to evaluate the new CE-IVD-marked versions of these techniques on a selected subset (n = 100) of the specimens.
From January 2008 to June 2009, 314 consecutive FFPE CRC patient tissues were subjected to KRAS mutation analysis after a histological confirmation of adenocarcinoma and presence of tumor cells in haematoxylin-eosin-stained slides. In addition, the tumor cell fraction was estimated and, when possible, tissue samples were manually microdissected with a scalpel to enrich for tumor cells before DNA extraction. The project was conducted in compliance with the Helsinki Declaration. The patient samples were used in accordance with the Swedish Biobank Legislation and Ethical Review Act (approval by Uppsala Ethical Review Board, reference number: 2004:M-281 and 2009-224), including appropriate procedures for informed consent.
Depending on the size of the tissue sample, one to ten sections (10 μm thick) from the FFPE block were used for genomic DNA extraction in an EZ1 workstation using the EZ1 DNA tissue kit and EZ1 DNA paraffin section card (QIAGEN GmbH, Hilden, Germany) according to the manufacturer's instructions. The quality and concentration of the extracted DNA was determined using a NanoDrop instrument (Thermo Scientific, Wilmington, DE).
Samples for the CE-IVD-marked PyroMark and TheraScreen assay comparison
Of the 314 samples, a subset of 100 consecutive samples was used for a direct comparison of the two CE-marked assays. Sample inclusion criteria for this part of the study were a DNA concentration of ≥10 ng/μl and a total amount of 440 ng DNA remaining after the original diagnostic KRAS analysis. The 100 samples were analysed for mutations in codons 12 or 13 using both methods in a single blind fashion. The sample code key was broken after all analyses were completed.
The initial Pyrosequencing analysis of the 314 samples was performed according to the manufacturer's recommendations for PyroMark Q24 KRAS and PyroMark Q24 KRAS v2.0 assays. Briefly, 2×10 ng of genomic DNA were used for analyses of KRAS codons 12/13 and 61 in 2×25 μl PCR reactions. Twenty μl of each PCR product was subjected to Pyrosequencing analysis using Streptavidin Sepharose High Performance (GE Healthcare, Uppsala, Sweden), PyroMark Gold Q96 reagents, PyroMark Q24 1.0.9 software, and a Q24 instrument (QIAGEN). The mutation analysis using the CE-IVD-marked PyroMark KRAS kit (QIAGEN) in the comparative study of the 100 samples was performed according to the PyroMark KRAS Kit handbook, version 1, June 2009. Briefly, 10 ng of genomic DNA was used for the initial 25 μl PCR reaction for codon 12 and 13 mutation analysis of each sample. Ten μl of the PCR product was subjected to the Pyrosequencing reaction. The quality thresholds for the mutational analysis were a required peak height of 30 relative light units (RLU) for "passed" quality and 10 RLU for "check" quality. Samples with an initial "check" status, or with an indicated mutation signal of 2-5%, were subjected to a second round of analysis performed in triplicates. In addition, samples that failed the initial PyroMark KRAS analysis were subjected to a second round of analysis.
Analysis using the DxS TheraScreen K-RAS Mutation Kit
The TheraScreen analysis (DxS Ltd, Manchester, UK) is an allele-specific PCR-based technology with specific primers for the seven most common KRAS codon 12 and 13 mutations. The assay screen for the following mutations: 12 GCT (Ala), 12 GAT (Asp), 12 CGT (Arg), 12 TGT (Cys), 12 AGT (Ser), 12 GTT (Val), and 13 GAC (Asp). Mutation analysis was performed according to instructions in the TheraScreen K-RAS Mutation Kit manual, version DU001e, January 2009 on an ABI PRISM® 7900HT SDS PCR instrument (Applied Biosystems Inc., Foster City, CA). In total 8×15 ng genomic DNA was used for the initial analysis of each sample. The quality thresholds when using the TheraScreen K-RAS Mutation Kit followed the recommendations in the manual, e.g. samples with a control assay with a cycle threshold (Ct) of 35 or higher should be rejected and samples with a mutation signal of Ct ≥38 should be scored as negative (wild-type).
DNA quality control analysis
Analysis of DNA quality for samples that gave suboptimal results in both assays was performed by a multiplex PCR reaction (Specimen Control Size Ladder, InVivoScribe Technologies, San Diego, CA) for amplification of five different DNA targets (100, 200, 300, 400 and 600 basepairs), according to the manufacturer's instructions. Products were visualised by electrophoresis on a 2% agarose gel.
Frequency and distribution of KRAS mutations
Tumor cell content and analytical sensitivity
Sensitivity of the Pyrosequencing mutation detection
Our diagnostic algorithm for KRAS analysis includes reanalysis of all samples with < 10% tumor cells using an independent assay. Thus, samples with a low tumor cell count (n = 56) were also analysed by a mutation specific PCR method (TheraScreen K-RAS Mutation kit). All codon 12/13 mutations detected by Pyrosequencing were confirmed, and in three additional samples indications of KRAS mutations were found. However, of these three samples, only one met the formal analysis criteria of the TheraScreen assay, i.e. that the real-time PCR signal for a mutation had a Ct value below 38. This sample had 5% tumor cells and a codon 12 GGT>GAT mutation.
Comparison of the CE-IVD-marked PyroMark and TheraScreen K-RAS assays
When performing clinical KRAS analyses, some samples will generate a borderline result due to low overall signal strength or a weak mutation signal. Validated assays, with a stringent definition of a true positive signal, are therefore essential. Since May 2009 there is a CE-marked Pyrosequencing kit (PyroMark KRAS kit, QIAGEN) available for in vitro diagnostic use in Europe. Therefore, we compared the performance of this new Pyrosequencing kit with the CE-marked TheraScreen K-RAS Mutation kit in a single blinded study protocol. DNA aliquots from a consecutive subset (n = 100) of the 314 CRC samples were analysed for KRAS mutations in codons 12 and 13 using both methods, and interpreted according to the manuals regarding procedures and criteria for positive and negative mutation results.
Eighty-nine of the 100 samples analysed with the PyroMark KRAS kit passed the quality criteria in the initial run. Four samples had an initial, and final, "failed" status (signal strength below the defined limits). Because of weak signal strength or indications of a KRAS mutation signal, the remaining seven samples were reanalysed in triplicates. After reanalysis, five of the samples passed the quality criteria. Thus, the mutational status of codons 12 and 13 could be defined for 94% of the samples by using this method.
Failed, rejected, and reanalysed samples in the comparison of the PyroMark and TheraScreen K-RAS assays
Pyro: Failed, repeated twice. DxS: Rejected, control Ct > 35
Pyro: Check, triplicate = wt. DxS: Rejected, control Ct > 35
Pyro: Check, triplicate = wt. DxS: Rejected, control Ct > 35
Pyro: Check, triplicate = inconclusive. DxS: Rejected, control Ct > 35
Pyro: Triplicate = wt. DxS: 12 GTT mutation
DxS: Rejected, control Ct > 35
Pyro: Failed. DxS: Rejected, control Ct > 35
Pyro: Check, triplicate = 12 GAT mutation. DxS: Rejected, control Ct > 35
Pyro: Failed, repeated in triplicate. DxS: Rejected, control Ct > 35
Pyro: Failed. DxS: Rejected, control Ct > 35
Pyro: Check, triplicate = inconclusive. DxS: triplicate = wt
Pyro: Low mutation signal, triplicate = 12 GTT mutation. DxS: 12 GTT mutation
Of the 94 samples, for which the mutational status could be defined using the PyroMark KRAS assay, 33 exhibited a KRAS mutation in codon 12 or 13. Similarly, 33 of the 91 passed samples in the TheraScreen mutational analysis were KRAS-mutated. When the DNA sample code was broken, 89 samples passed both assays and had the same mutational status. Five samples (DNA# 21, 37, 64, 67 and 89) failed the quality criteria of both assays (Table 1). Four samples (DNA# 25, 30, 61 and 66), were rejected in the TheraScreen assay, but could be defined as KRAS wt or codon 12 mutated (DNA# 66) when analysed by Pyrosequencing. Conversely, one sample (DNA# 93) gave inconclusive results in the Pyrosequencing assay but was scored as KRAS wt by the TheraScreen assay. Finally, DNA# 54 was defined as 12Val-mutated by TheraScreen analysis, but this mutation could not be unequivocally confirmed in the PyroMark assay.
Evaluation of DNA quality in rejected samples
Mutated RAS was first identified by its ability to transform cell lines in DNA transfection experiments. It was then concluded that the transforming ability was due to activating point mutations resulting in an amino acid substitution at codon 12, 13, or 61 [13, 29, 30]. However, mutation analysis of codon 61, or rare mutations in codon 12 or 13, were not included in the clinical studies that lie behind the recommendation to use anti-EGFR therapies only for KRAS wt metastatic CRC [31, 32].
In our cohort of 314 colorectal cancer patients, 39% (n = 122) of the samples were mutated in KRAS codon 12, 13 or 61. As expected, most mutations were located in codon 12 or 13 (n = 112) and represented well-established types of mutations in CRC . One unusual codon 12 mutation (GGT > TTT) was also identified. This mutation has, to our knowledge, only been reported three times [2, 33–35]. The frequency of codon 61 mutations is generally believed to be low. However, similar to a recent study , 8% (n = 10) of the detected KRAS mutations in our cohort were located in codon 61. Four of these result in a change of the wt glutamic acid to histidine. In the infrequent, but highly malignant, signet ring cell CRC KRAS codon 12 and 13 mutations are less frequent than in other CRC subtypes . Interestingly, more codon 61 histidine substitutions than codon 12 or 13 mutations have been reported in signet ring cell CRC . Although the distribution and relative frequencies of the mutation types can vary between different populations of CRC patients (reviewed by Packham et al. ), we argue that codon 61 mutations are more common than generally believed. The reason for the low incidence of codon 61 mutations reported in the Catalogue of somatic mutations in cancer database (0.3%)  and elsewhere (0-1.6%) [14–18] might be due to the use of sub-optimal sequencing methods or, simply, the widespread use of methods limited to codon 12 and 13 analysis. The clinical implication of KRAS codon 61 mutations in the treatment of CRC patients with EGFR inhibitors is not fully understood. However, a recent study indicates that codon 61 mutations are associated with resistance to cetuximab treatment .
The analytical sensitivity and specificity to detect a KRAS mutation in a clinical CRC sample will depend on (i) the technical ability to detect mutated alleles in a background of normal wt alleles, (ii) the sensitivity to DNA degradation in terms of DNA template target lengths, and (iii) the defined thresholds and signal strengths required for a positive or negative mutation call. The presence of normal cells, such as infiltrating inflammatory cells and stromal cells, will reduce the possibility to detect cancer-associated somatic mutations in a tumor sample. In clinical practice, even after manual microdissection, samples with a low tumor cell fraction are relatively common. In our series 18% of the samples had a final tumor cell content of < 10%. For these samples direct Sanger sequencing will not be sensitive enough as a tumor cell fraction of at least 25-30% is believed to be required for mutation detection by this method . Hence, a more sensitive method is needed for an accurate analysis of these samples. Previous studies using either in-house developed, or PyroMark v2.0, Pyrosequencing assays have defined a detection limit of 5% mutant alleles [22, 38]. According to the TheraScreen (DxS) manual, this kit has a reported sensitivity of 1% (of mutated alleles among wt alleles). This would correspond to a tumor cell content of 2% when detecting a heterozygote mutation in the tumor cells. Here we report that a similar sensitivity can be reached using the Pyrosequencing kit. We believe that the titration experiment performed in this study, using tumor and normal DNA from FFPE tissue, possibly better reflects the true diagnostic situation compared to dilution experiments with recombinant DNA or mixtures of non-fixed cultured cell lines. However, we find that both assays are sensitive to poor DNA quality. Indeed, DNA fragmentation induced by formalin fixation and/or degradation due to necrosis will reduce the success rate of PCR amplification, and might potentially introduce mutation artefacts [39, 40].
In a blinded sub-study of 100 CRC samples, we compared the two CE-marked mutation detection kits, PyroMark KRAS and TheraScreen K-RAS mutation kit, by adhering to the manufacturers' protocols and guidelines for interpretation of the results. The mutational status could be defined for 94 and 91 samples, respectively, and both assays detected the same number of mutated samples. After analysis of DNA fragmentation, we concluded that suboptimal DNA quality was the reason that some samples did not pass the quality thresholds. Analysis of such samples resulted in a high control Ct (TheraScreen) or low signal strength (PyroMark). Both assays uncovered identical mutations in 32 of the samples. In addition, one sample that was rejected in the TheraScreen assay was found KRAS mutated when analysed with the PyroMark assay. Conversely, a mutation defined by the TheraScreen assay in one sample could not be exclusively confirmed by the PyroMark assay. Our experience when comparing the two assays in clinical practice is that the TheraScreen assay is possibly slightly more sensitive in terms of detecting low numbers of mutated alleles. However, as this assay is dependent on several real-time PCR reactions in combination with a strict regulatory definition of true mutations and thresholds for rejection (in terms of Ct values), this assay seems to be more dependent on high DNA quality in comparison to the PyroMark assay. Moreover, less DNA is needed for the PyroMark assay (10 ng) in comparison with the TheraScreen method (8×2-20 ng). The amount of DNA required for a mutation analysis can be of importance when analysing minimal colonoscopy fragments or core needle biopsies. Similar to our DxS TheraScreen results, a failure rate of 5-10% have been reported when analysing CRC samples for KRAS mutations using the DxS technology. The failed samples either had limited material or poor quality of the DNA [41, 42]. In comparison, when analysed with Pyrosequencing, the failure rate was 2.5% for the 314 consecutive CRC samples in this study. Finally, we want to emphasise that the criteria for "mutations" or "rejections", as defined by manufacturers of kits, or individual molecular pathology laboratories, adds a subjective dimension to the interpretation of the results that goes beyond a pure technical comparison of different assays.
Our conclusion from the KRAS analysis of the 314 clinical samples together with data from the comparison of the two CE-marked assays is that both assays are well suited for KRAS mutation analysis of DNA from FFPE CRC samples. Both assays have a sufficient analytical sensitivity to efficiently detect KRAS mutations even in samples with < 10% tumor cells.
The comparison of the PyroMark KRAS assay with TheraScreen K-RAS Mutation kit was financially supported by an unrestricted grant from QIAGEN GmbH. A short summary of their interpretation of the results can be found in the PyroMark KRAS Kit Handbook version 1, 2009.
- Andreyev HJ, Norman AR, Cunningham D, Oates J, Dix BR, Iacopetta BJ, Young J, Walsh T, Ward R, Hawkins N, et al: Kirsten ras mutations in patients with colorectal cancer: the 'RASCAL II' study. Br J Cancer. 2001, 85 (5): 692-696. 10.1054/bjoc.2001.1964.View ArticlePubMedPubMed CentralGoogle Scholar
- Cosmic. [http://www.sanger.ac.uk/genetics/CGP/cosmic]
- Goncalves A, Esteyries S, Taylor-Smedra B, Lagarde A, Ayadi M, Monges G, Bertucci F, Esterni B, Delpero JR, Turrini O, et al: A polymorphism of EGFR extracellular domain is associated with progression free-survival in metastatic colorectal cancer patients receiving cetuximab-based treatment. BMC Cancer. 2008, 8: 169-10.1186/1471-2407-8-169.View ArticlePubMedPubMed CentralGoogle Scholar
- Amado RG, Wolf M, Peeters M, Van Cutsem E, Siena S, Freeman DJ, Juan T, Sikorski R, Suggs S, Radinsky R, et al: Wild-type KRAS is required for panitumumab efficacy in patients with metastatic colorectal cancer. J Clin Oncol. 2008, 26 (10): 1626-1634. 10.1200/JCO.2007.14.7116.View ArticlePubMedGoogle Scholar
- Karapetis CS, Khambata-Ford S, Jonker DJ, O'Callaghan CJ, Tu D, Tebbutt NC, Simes RJ, Chalchal H, Shapiro JD, Robitaille S, et al: K-ras mutations and benefit from cetuximab in advanced colorectal cancer. N Engl J Med. 2008, 359 (17): 1757-1765. 10.1056/NEJMoa0804385.View ArticlePubMedGoogle Scholar
- Van Cutsem E, Kohne CH, Hitre E, Zaluski J, Chang Chien CR, Makhson A, D'Haens G, Pinter T, Lim R, Bodoky G, et al: Cetuximab and chemotherapy as initial treatment for metastatic colorectal cancer. N Engl J Med. 2009, 360 (14): 1408-1417. 10.1056/NEJMoa0805019.View ArticlePubMedGoogle Scholar
- Lievre A, Bachet JB, Le Corre D, Boige V, Landi B, Emile JF, Cote JF, Tomasic G, Penna C, Ducreux M, et al: KRAS mutation status is predictive of response to cetuximab therapy in colorectal cancer. Cancer Res. 2006, 66 (8): 3992-3995. 10.1158/0008-5472.CAN-06-0191.View ArticlePubMedGoogle Scholar
- Lievre A, Bachet JB, Boige V, Cayre A, Le Corre D, Buc E, Ychou M, Bouche O, Landi B, Louvet C, et al: KRAS mutations as an independent prognostic factor in patients with advanced colorectal cancer treated with cetuximab. J Clin Oncol. 2008, 26 (3): 374-379. 10.1200/JCO.2007.12.5906.View ArticlePubMedGoogle Scholar
- De Roock W, Piessevaux H, De Schutter J, Janssens M, De Hertogh G, Personeni N, Biesmans B, Van Laethem JL, Peeters M, Humblet Y, et al: KRAS wild-type state predicts survival and is associated to early radiological response in metastatic colorectal cancer treated with cetuximab. Ann Oncol. 2008, 19 (3): 508-515. 10.1093/annonc/mdm496.View ArticlePubMedGoogle Scholar
- Benvenuti S, Sartore-Bianchi A, Di Nicolantonio F, Zanon C, Moroni M, Veronese S, Siena S, Bardelli A: Oncogenic activation of the RAS/RAF signaling pathway impairs the response of metastatic colorectal cancers to anti-epidermal growth factor receptor antibody therapies. Cancer Res. 2007, 67 (6): 2643-2648. 10.1158/0008-5472.CAN-06-4158.View ArticlePubMedGoogle Scholar
- Velho S, Moutinho C, Cirnes L, Albuquerque C, Hamelin R, Schmitt F, Carneiro F, Oliveira C, Seruca R: BRAF, KRAS and PIK3CA mutations in colorectal serrated polyps and cancer: primary or secondary genetic events in colorectal carcinogenesis?. BMC Cancer. 2008, 8: 255-10.1186/1471-2407-8-255.View ArticlePubMedPubMed CentralGoogle Scholar
- Packham D, Ward RL, Ap Lin V, Hawkins NJ, Hitchins MP: Implementation of novel pyrosequencing assays to screen for common mutations of BRAF and KRAS in a cohort of sporadic colorectal cancers. Diagnostic Molecular Pathology. 2009, 18 (2): 62-71. 10.1097/PDM.0b013e318182af52.View ArticlePubMedGoogle Scholar
- Bos JL: ras oncogenes in human cancer: a review. Cancer Res. 1989, 49 (17): 4682-4689.PubMedGoogle Scholar
- Poehlmann A, Kuester D, Meyer F, Lippert H, Roessner A, Schneider-Stock R: K-ras mutation detection in colorectal cancer using the Pyrosequencing technique. Pathol Res Pract. 2007, 203 (7): 489-497. 10.1016/j.prp.2007.06.001.View ArticlePubMedGoogle Scholar
- Chang SC, Lin JK, Yang SH, Wang HS, Li AF, Chi CW: Relationship between genetic alterations and prognosis in sporadic colorectal cancer. Int J Cancer. 2006, 118 (7): 1721-1727. 10.1002/ijc.21563.View ArticlePubMedGoogle Scholar
- Chang YS, Yeh KT, Chang TJ, Chai C, Lu HC, Hsu NC, Chang JG: Fast simultaneous detection of K-RAS mutations in colorectal cancer. BMC Cancer. 2009, 9: 179-10.1186/1471-2407-9-179.View ArticlePubMedPubMed CentralGoogle Scholar
- Edkins S, O'Meara S, Parker A, Stevens C, Reis M, Jones S, Greenman C, Davies H, Dalgliesh G, Forbes S, et al: Recurrent KRAS codon 146 mutations in human colorectal cancer. Cancer Biol Ther. 2006, 5 (8): 928-932. 10.4161/cbt.5.8.3251.View ArticlePubMedPubMed CentralGoogle Scholar
- Smith G, Carey FA, Beattie J, Wilkie MJ, Lightfoot TJ, Coxhead J, Garner RC, Steele RJ, Wolf CR: Mutations in APC, Kirsten-ras, and p53--alternative genetic pathways to colorectal cancer. Proc Natl Acad Sci USA. 2002, 99 (14): 9433-9438. 10.1073/pnas.122612899.View ArticlePubMedPubMed CentralGoogle Scholar
- Richman SD, Seymour MT, Chambers P, Elliott F, Daly CL, Meade AM, Taylor G, Barrett JH, Quirke P: KRAS and BRAF mutations in advanced colorectal cancer are associated with poor prognosis but do not preclude benefit from oxaliplatin or irinotecan: results from the MRC FOCUS trial. J Clin Oncol. 2009, 27 (35): 5931-5937. 10.1200/JCO.2009.22.4295.View ArticlePubMedGoogle Scholar
- Loupakis F, Ruzzo A, Cremolini C, Vincenzi B, Salvatore L, Santini D, Masi G, Stasi I, Canestrari E, Rulli E, et al: KRAS codon 61, 146 and BRAF mutations predict resistance to cetuximab plus irinotecan in KRAS codon 12 and 13 wild-type metastatic colorectal cancer. British Journal of Cancer. 2009, 101 (4): 715-721. 10.1038/sj.bjc.6605177.View ArticlePubMedPubMed CentralGoogle Scholar
- Ronaghi M, Uhlen M, Nyren P: A sequencing method based on real-time pyrophosphate. Science. 1998, 281 (5375): 363, 365-10.1126/science.281.5375.363.View ArticlePubMedGoogle Scholar
- Ogino S, Kawasaki T, Brahmandam M, Yan L, Cantor M, Namgyal C, Mino-Kenudson M, Lauwers GY, Loda M, Fuchs CS: Sensitive sequencing method for KRAS mutation detection by Pyrosequencing. J Mol Diagn. 2005, 7 (3): 413-421.View ArticlePubMedPubMed CentralGoogle Scholar
- Zuo Z, Chen SS, Chandra PK, Galbincea JM, Soape M, Doan S, Barkoh BA, Koeppen H, Medeiros LJ, Luthra R: Application of COLD-PCR for improved detection of KRAS mutations in clinical samples. Mod Pathol. 2009, 22 (8): 1023-1031. 10.1038/modpathol.2009.59.View ArticlePubMedGoogle Scholar
- Cross J: DxS Ltd. Pharmacogenomics. 2008, 9 (4): 463-467. 10.2217/146224220.127.116.113.View ArticlePubMedGoogle Scholar
- Kotoula V, Charalambous E, Biesmans B, Malousi A, Vrettou E, Fountzilas G, Karkavelas G: Targeted KRAS mutation assessment on patient tumor histologic material in real time diagnostics. PLoS One. 2009, 4 (11): e7746-10.1371/journal.pone.0007746.View ArticlePubMedPubMed CentralGoogle Scholar
- Hecht JR, Mitchell E, Chidiac T, Scroggin C, Hagenstad C, Spigel D, Marshall J, Cohn A, McCollum D, Stella P, et al: A randomized phase IIIB trial of chemotherapy, bevacizumab, and panitumumab compared with chemotherapy and bevacizumab alone for metastatic colorectal cancer. J Clin Oncol. 2009, 27 (5): 672-680. 10.1200/JCO.2008.19.8135.View ArticlePubMedGoogle Scholar
- Tol J, Koopman M, Cats A, Rodenburg CJ, Creemers GJ, Schrama JG, Erdkamp FL, Vos AH, van Groeningen CJ, Sinnige HA, et al: Chemotherapy, bevacizumab, and cetuximab in metastatic colorectal cancer. N Engl J Med. 2009, 360 (6): 563-572. 10.1056/NEJMoa0808268.View ArticlePubMedGoogle Scholar
- Micke P, Ostman A: Exploring the tumour environment: cancer-associated fibroblasts as targets in cancer therapy. Expert Opin Ther Targets. 2005, 9 (6): 1217-1233. 10.1517/1472818.104.22.1687.View ArticlePubMedGoogle Scholar
- Bos JL, Fearon ER, Hamilton SR, Verlaan-de Vries M, van Boom JH, van der Eb AJ, Vogelstein B: Prevalence of ras gene mutations in human colorectal cancers. Nature. 1987, 327 (6120): 293-297. 10.1038/327293a0.View ArticlePubMedGoogle Scholar
- Der CJ, Finkel T, Cooper GM: Biological and biochemical properties of human rasH genes mutated at codon 61. Cell. 1986, 44 (1): 167-176. 10.1016/0092-8674(86)90495-2.View ArticlePubMedGoogle Scholar
- EMEA: EMEA/405113/2007. 2007Google Scholar
- EMEA: EMEA/CHMP/280402/2008. 2008Google Scholar
- Nusko G, Sachse R, Mansmann U, Wittekind C, Hahn EG: K-RAS-2 gene mutations as predictors of metachronous colorectal adenomas. Scand J Gastroenterol. 1997, 32 (10): 1035-1041. 10.3109/00365529709011221.View ArticlePubMedGoogle Scholar
- Tortola S, Steinert R, Hantschick M, Peinado MA, Gastinger I, Stosiek P, Lippert H, Schlegel W, Reymond MA: Discordance between K-ras mutations in bone marrow micrometastases and the primary tumor in colorectal cancer. J Clin Oncol. 2001, 19 (11): 2837-2843.PubMedGoogle Scholar
- Neumann J, Zeindl-Eberhart E, Kirchner T, Jung A: Frequency and type of KRAS mutations in routine diagnostic analysis of metastatic colorectal cancer. Pathol Res Pract. 2009, 205 (12): 858-862. 10.1016/j.prp.2009.07.010.View ArticlePubMedGoogle Scholar
- Ogino S, Brahmandam M, Cantor M, Namgyal C, Kawasaki T, Kirkner G, Meyerhardt JA, Loda M, Fuchs CS: Distinct molecular features of colorectal carcinoma with signet ring cell component and colorectal carcinoma with mucinous component. Mod Pathol. 2006, 19 (1): 59-68. 10.1038/modpathol.3800482.View ArticlePubMedGoogle Scholar
- Wistuba II, Behrens C, Albores-Saavedra J, Delgado R, Lopez F, Gazdar AF: Distinct K-ras mutation pattern characterizes signet ring cell colorectal carcinoma. Clin Cancer Res. 2003, 9 (10 Pt 1): 3615-3619.PubMedGoogle Scholar
- Tsiatis AC, Norris-Kirby A, Rich RG, Hafez MJ, Gocke CD, Eshleman JR, Murphy KM: Comparison of Sanger sequencing, pyrosequencing, and melting curve analysis for the detection of KRAS mutations: diagnostic and clinical implications. J Mol Diagn. 2010, 12 (4): 425-432. 10.2353/jmoldx.2010.090188.View ArticlePubMedPubMed CentralGoogle Scholar
- Ben-Ezra J, Johnson DA, Rossi J, Cook N, Wu A: Effect of fixation on the amplification of nucleic acids from paraffin-embedded material by the polymerase chain reaction. J Histochem Cytochem. 1991, 39 (3): 351-354.View ArticlePubMedGoogle Scholar
- Williams C, Ponten F, Moberg C, Soderkvist P, Uhlen M, Ponten J, Sitbon G, Lundeberg J: A high frequency of sequence alterations is due to formalin fixation of archival specimens. Am J Pathol. 1999, 155 (5): 1467-1471.View ArticlePubMedPubMed CentralGoogle Scholar
- Garm Spindler KL, Pallisgaard N, Rasmussen AA, Lindebjerg J, Andersen RF, Cruger D, Jakobsen A: The importance of KRAS mutations and EGF61A>G polymorphism to the effect of cetuximab and irinotecan in metastatic colorectal cancer. Ann Oncol. 2009, 20 (5): 879-884. 10.1093/annonc/mdn712.View ArticleGoogle Scholar
- Whitehall V, Tran K, Umapathy A, Grieu F, Hewitt C, Evans TJ, Ismail T, Li WQ, Collins P, Ravetto P, et al: A multicenter blinded study to evaluate KRAS mutation testing methodologies in the clinical setting. J Mol Diagn. 2009, 11 (6): 543-552. 10.2353/jmoldx.2009.090057.View ArticlePubMedPubMed CentralGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2407/10/660/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.